Journal:Development of a gas-chromatographic method for simultaneous determination of cannabinoids and terpenes in hemp
|Full article title||Development of a gas-chromatographic method for simultaneous determination of cannabinoids and terpenes in hemp|
|Author(s)||Zekič, Jure; Križman, Mitja|
|Author affiliation(s)||National Institute of Chemistry - Ljubljana, University of Ljubljana|
|Primary contact||Email: mitja dot krizman at ki dot si|
|Volume and issue||25(24)|
|Distribution license||Creative Commons Attribution 4.0 International|
An original gas-chromatographic method has been developed for simultaneous determination of major terpenes and cannabinoids in plant samples and their extracts. The main issues to be addressed were not only the large differences in polarity and volatility between both groups of analytes, but also the need for an exhaustive decarboxylation of cannabinoid acidic forms. Sample preparation was minimized by avoiding any analyte derivatization. Acetone was found to be the most appropriate extraction solvent. Successful chromatographic separation was achieved by using a medium-polarity column. Limits of detection ranged from 120 to 260 ng/mL for terpenes and from 660 to 860 ng/mL for cannabinoids. Parallel testing proved the results for cannabinoids are comparable to those obtained from established high-performance liquid chromatography (HPLC) methods. Despite very large differences in concentrations between both analyte groups, a linear range between 1 and 100 µg/mL for terpenes and between 10 and 1500 µg/mL for cannabinoids was determined.
Keywords: cannabinoids, terpenes, cannabis, hemp, gas chromatography, capillary column
The hemp plant (Cannabis sativa] and Cannabis indica), or simply Cannabis, is a plant that has elicited much interest throughout history because of its characteristics and various possibilities of use. Over the last few years, the popularity of the Cannabis plant and its constituents has particularly increased, and a widespread recognition of its usefulness, including for medical purposes, is becoming increasingly noticeable. Hemp is known to contain various groups of compounds, probably the most characteristic among them being cannabinoids. Furthermore, cannabis also contains a diverse array of terpenes and flavonoids, as well as other groups of compounds.
Cannabinoids are probably the most studied metabolites of cannabis. Many of their beneficial effects on human health are already known, and there is also a lot of ongoing research, discovering new ones. As a result, the use of cannabinoids in a wide variety of preparations is growing, which is also reflected in increased cannabis production. At the same time, a need for an efficient, routine analytical method for monitoring the cannabinoid content in plant material has arisen. A number of methods for the analysis of cannabinoids in cannabis have indeed already been developed; among various approaches, the predominant is chromatographic analysis, in particular using gas chromatography (GC) or high-performance liquid chromatography (HPLC).
Even though gas chromatography used to be the most common technique for analysis of cannabinoids in cannabis extracts, HPLC is currently increasingly gaining popularity in this field of application. HPLC determination of cannabinoids, in comparison to the analysis with GC, has some significant advantages: above all, it avoids the potential aggravating circumstances caused by the high temperature of analysis associated with GC, which affects the results mainly during the phase of sample injection and also indirectly during the analysis itself. Cannabinoids are found mainly in acidic forms in the plant, which eventually decarboxylate if they are exposed to raised temperature. The temperature in the gas chromatograph also causes the process of decarboxylation, which is reflected in the results in two ways: we cannot separately determine acidic and decarboxylated forms of a particular cannabinoid, but only their total content. On the other hand, there is a significant probability that decarboxylation in the injector will not proceed completely. Especially at higher cannabinoid concentrations, this may be reflected in apparently lower values measured and consequently irregular results of analysis. Both problems can be successfully solved by the derivatization of cannabinoids (including their acid forms) in the sample. However, this represents an additional step that is often not desirable, because it increases probability for experimental error and prolongs analysis time, which may be a considerable drawback in terms of method suitability for routine use. With HPLC, all of these problems have been successfully avoided, as some relatively rapid, simple, and effective methods for the determination of both acidic and decarboxylated cannabinoids in cannabis samples have already been developed.
Thus, two major approaches to chromatographic analysis of cannabinoids in hemp most often appear in the literature; direct analysis of a suitably diluted sample extract by liquid chromatography, or preliminary derivatization of the extract and subsequent analysis by gas chromatography. Despite its mentioned drawbacks, the latter approach is still quite in use, somewhat for traditional reasons, but also for entirely practical reasons, since GC instrumentation is simpler and less expensive than HPLC, more economical for use, or maybe even the only option available.
For direct gas chromatographic analysis of cannabinoids, traditionally, the most commonly used stationary phase is 5% phenyl 95% dimethylpolysiloxane, followed by 100% dimethylpolysiloxane phase. Recently, more polar stationary phases like 35% phenyl 65% dimethylpolysiloxane have also been used, with a potential gain in selectivity.
Another relatively important group of compounds in cannabis are terpenes. Different varieties of cannabis contain mainly different monoterpenes and sesquiterpenes, which also give a distinctive scent to hemp plants. From an analytical point of view, they are interesting because their profile is often characteristic of a particular variety or population of cannabis, which may enable identification of different plant specimens. Terpenes in cannabis are also often credited for the so-called “entourage” effect. The analysis of terpenes is most often performed using gas chromatography as a separate type of analysis; successful separation and determination on different types of columns is usually quite fast, effective, and simple.
An unavoidable step in chromatographic analysis of plant material is analyte extraction from the sample. Both groups of compounds, terpenes and cannabinoids, can be extracted from the plant material by different approaches. For cannabinoids, the most common is classical extraction with a relatively apolar solvent (usually ethanol) either by mechanical shaking or by ultrasonic extraction. On the other hand, GC analysis of terpenes can also be done by the headspace sampling technique. The alternative to headspace sampling is of course solvent extraction, in such a case a solvent of appropriate polarity must be chosen in relation to the analytes of interest. Terpenes and cannabinoids differ both in terms of volatility and polarity, as well as in the concentrations found in hemp samples. Terpene levels are usually significantly lower compared to cannabinoids. As demonstrated by Namdar et al., an optimum solvent for terpene extraction was found to be a mixture of ethanol:hexane (3:7, v/v), while for cannabinoids they corroborated the use of ethanol as the optimum solvent.
Given the above, a simultaneous analysis of terpenes and cannabinoids in hemp samples is not a trivial task. A notable example of such a type of methodology has been published by Franchina et al. In that case, the methodology involved the use of sorptive extraction and thermal desorption sampling, two-dimensional gas chromatography, and mass detection. Such a methodology is certainly very detailed and useful when advanced studies have to be done, like in chemotaxonomy. For routine analyses, however, such a setup is probably too complicated and expensive.
The aim of this work was to find appropriate conditions, mainly in terms of sample preparation, for a simultaneous analysis of both groups of compounds, while keeping the overall experimental and instrumental setups simple.
Results and discussion
As already stated in the introduction, the main challenge in combined analysis of terpenes and cannabinoids is the sample preparation step, and more precisely the extraction conditions. As Namdar et al. have noted, optimum solvent composition for terpenes and cannabinoids differs. During their work, they found the mixture of ethanol:hexane (3:7, v/v) to be the best compromise for extracting both groups of compounds. In this present work, the quest for a single solvent similar in properties to the mentioned work was undertaken. The solvent selection was then narrowed to acetone and ethyl acetate, with the added benefit of them being solvents with low environmental impact. Finally, acetone was selected as the most appropriate solvent, based on the extraction recoveries obtained.
Another important parameter highly affecting the results is the sample-to-volume ratio during extraction. Besides having a direct effect on extraction efficiency as well, this ratio also has implications on the final analyte concentrations. Good overall analyte recoveries were obtained with sample-to-volume ratios between 1:10 and 1:25. During method development, it was found that a ratio of about 1:17 (i.e., 300 mg per 5 mL of solvent) was a good compromise between extraction efficiency while at the same time still providing sufficient concentrations of terpenes in working sample solutions in order to be quantified without concentrating the solution. Terpenes are unfortunately very volatile, and significant analyte losses can be expected with any of the solvent evaporation techniques. Therefore, this sample preparation step was deliberately avoided. At the same time, cannabinoid concentrations in sample solutions proved to be below the upper practical limit in terms of detector linearity. Compared to terpenes, cannabinoids are more problematic to analyze. Besides their lower volatility, the possibility of incomplete decarboxylation of cannabinoid acidic forms during sample vaporization and injection must be prevented. This phenomenon is more pronounced at higher concentrations.
In practice, the concentrations of major cannabinoids in the extracts were kept at up to 1.5 mg/mL or below. By using such high cannabinoid concentrations, it was less challenging to quantify minor cannabinoids as well.
Gas chromatographic separation
Terpenes and cannabinoids differ widely, both in terms of polarity and volatility. These two groups of compounds are therefore easily separated between each other using GC, although a wide temperature gradient program is needed due to a large difference in volatility. Successful separation of individual terpenes is not particularly challenging, as many works demonstrate. The main challenge was to provide good separation of some cannabinoids, the most critical being the resolution between cannabichromene (CBC) and cannabidiol (CBD). Using the ubiquitous stationary phase based on 5% phenyl 95% dimethylpolysiloxane, the resolution between those two peaks proved to be unsuitable, since these two peaks overlapped, as proven by preliminary tests (data not published). Much better results are obtained using more polar stationary phases like 35% phenyl 65% dimethylpolysiloxane, as recent applications also demonstrate. As a consequence, the choice for an even more polar stationary phase was made, namely 50% phenyl 50% dimethylpolysiloxane. According to initial expectations, a good resolution was obtained, and no issues related to overlapping of cannabinoid peaks were observed anymore. At the same time, using a relatively polar stationary phase did not impair the separation of terpenes. In fact, even more polar stationary phases were employed for terpenes. Chromatograms of standard solutions and sample extracts are depicted on Figure 1, Figure 2. and Figure 3.
Method performance and validation
The developed method exhibited good overall analytical performance within a relatively short analysis time, since it provided separation of two major groups of compounds in the cannabis plant. The most difficult to separate, CBD and CBC, were fully baseline resolved. Other analytes were also separated with excellent resolution. Validation and stability data (Table 1) confirmed the suitability of the method also for quantitative use. Sufficiently low sensitivity limits, which are important (especially from the terpenes standpoint), while also maintaining accurate results for higher cannabinoid concentrations, allow for the use of non-concentrated or non-diluted working sample solutions. This is a great advantage, especially in view of the process' simplicity.
Comparison with HPLC
In order to confirm correctness of the results obtained from the newly developed GC method, HPLC analysis was performed, applying a previously published and validated method. As mentioned before, GC analysis may not be sufficiently accurate mainly at higher cannabinoid concentrations. Therefore, HPLC quantification was performed for the two major cannabinoids in the samples—cannabidiol or cannabigerol—and their acidic forms which occur in rather high concentrations. The total content of these cannabinoids, measured by HPLC analysis, was in very high accordance with results obtained from GC analysis (Table 2). This also directly confirms the correctness of the developed GC method in this respect. Aside from being inherently a less sensitive methodology than HPLC, as depicted in Table 1, GC cannabinoid analysis is performance-wise fully comparable to HPLC. For most analytes, better repeatability figures were also obtained.
Materials and methods
Chemicals, reagents, and samples
Practical grade ethyl acetate and n-hexane from Merck (Darmstadt, Germany) and acetone from Honeywell (Seelze, Germany) were used for standards preparation and sample treatment. HPLC grade methanol from Honeywell (Seelze, Germany) and deionised water from a Milli-Q apparatus (Millipore, Milford, MA, USA) were used to prepare mobile phase for HPLC analysis. Ammonium formate was LC–MS grade (Sigma-Aldrich, St. Louis, MO, USA).
Terpene reference standards of α-humulene (96%), limonene (97%), myrcene (>90%), α-pinene (99%), β-pinene (97%), and α-terpinene (95%) were obtained from Sigma-Aldrich (St. Louis, MO, USA).
Cannabinoid reference standards of cannabigerol (CBG, 99%) and cannabidiol (CBD, 99.9%) in solid form were obtained from LGC standards (Teddington, Middlesex, UK). Δ8-tetrahydrocannabinol (Δ8-THC), Δ9-tetrahydrocannabinol (Δ9-THC), cannabichromene (CBC), and cannabinol (CBN) were obtained as solution in methanol (1 mg/mL) from LGC standards as well.
Squalane (96%) was purchased from Sigma-Aldrich (St. Louis, MO, USA) and was applied as internal standard (IS).
Buds of industrial hemp plant (stemless) material of various cultivars were obtained from local hemp growers.
Analysis was performed by a Focus GC with a flame ionization detector (FID) (Thermo Scientific, Rodano, Milan, Italy) with a Rtx-50 capillary column (30 m × 0.25 mm × 0.25 µm film thickness) (Restek Corporation, Bellefonte, U.S.A.). One microlitre of sample was injected with a split ratio 30:1 at 310 °C using ultrapure grade helium as carrier gas at 2 mL/min. The GC oven temperature program started at 60 °C (3 min), followed by a linear gradient of 20 °C/min to 290 °C, temperature was then kept constant for 8 min. Flame ionisation detector temperature was 310 °C.
Thermo Electron Trace 2000 GC coupled with Thermo Electron DSQ quadrupole mass spectrometer (Thermo Scientific, Rodano, Milan, Italy) with an electron ionisation source at 70 eV in positive mode was applied for the purpose of terpene identification. Some parameters had to be adapted for gas chromatography–mass spectrometry (GC-MS) analysis. Namely, 0.5 µL of sample was injected with a split ratio of 60:1, flow rate of carrier gas was 1 mL/min. Other parameters remained the same as for GC-FID analysis. The MS data were acquired in full scan mode from m/z 50–450 with acquisition frequency of 4.2 scans per second.
Analysis was performed according to Križman's published method. For HPLC analysis, sample solutions (as described below in "Samples and preparation of extracts") were further diluted with methanol 100-fold.
Standard solutions preparation
Stock solutions of solid cannabinoid standards (cannabidiol and cannabigerol) were prepared in acetone in the concentration of 2.0 mg/mL. Working standard solutions of cannabidiol and cannabigerol were prepared in the concentration range of 0.05 to 1.5 mg/mL, and working standard solutions of cannabichromene, cannabinol, Δ8-tetrahydrocannabinol, and Δ9-tetrahydrocannabinol were prepared in the concentration range of 0.01 to 0.2 mg/mL. All working solutions contained internal standard of concentration 0.13 mg/mL and were prepared by diluting stock solution with acetone.
Stock solutions of terpene standards were prepared in the concentration of 2.0 mg/mL in acetone, except for myrcene stock solution, which was prepared in hexane. Working standard solutions consisted of a mix of all terpene standards in the concentration range of 1.0 to 100 µg/mL.
Samples and preparation of extracts
Dried and powdered plant materials (300 mg) were extracted by sonication for 30 min at room temperature with acetone or ethyl acetate containing IS (130 µg/mL). Sample solutions were then centrifuged at 16.000g for 10 min and the supernatant was transferred into GC vials.
Quantitation, method precision, accuracy, sensitivity, linearity, and stability
Injection precision was determined by five injections of working standard solution. Extraction efficiency was assessed by three consecutive extractions of selected plant samples, and then comparing the analyte recovery with the combined recovery of all extraction steps. Accuracy was determined by spiking sample solution with cannabinoid standards at concentration of 0.25 mg/mL and terpene standards at concentration of 0.025 mg/mL. Repeatability and intermediate precision were also tested on a homogeneous plant sample. Three replicates were assayed for repeatability, while three replicates were assayed on each of the three consecutive days for intermediate precision.
Linearity was checked in the range of standard solutions concentration (0.05 to 1.5 mg/mL for CBD and CBG; 0.01 to 0.2 mg/mL for CBC, CBN, Δ8-THC, and Δ9-THC; and 0.001 to 0.1 mg/mL for terpenes). Correlation coefficients were calculated with intercept values set at zero. Analyte peak areas were normalised by dividing them with IS peak areas. For stability tests, a sample solution was refrigerated at 4 °C in the dark for 48 h.
In order to confirm the correctness of the results, HPLC analysis of properly diluted sample extracts solutions in methanol was performed.
A gas chromatography–flame ionization detection (GC-FID) method for simultaneous analysis of terpenes and cannabinoids in hemp samples has been developed. The main issues concerning the method were ensuring appropriate sample preparation conditions for both terpenes and cannabinoids, successful separation of critical cannabinoid peaks, and quantitative decarboxylation of cannabinoid acidic forms. Acetone proved to be an appropriate solvent for quantitative extraction of all the analytes concerned, which occur in a wide concentration range, using a sufficiently low sample-to-solvent ratio. Separation-wise, the resolution between CBD and CBC peaks was substantially improved by using a relatively polar column with 50% phenyl 50% dimethylpolysiloxane stationary phase. Quantitative decarboxylation was ensured using a high injector temperature and low injection volumes. Peak identity was confirmed by GC-MS. Despite many of the method parameters being near the practical limits in terms of instrumentation capacity like temperature, detector response, etc., the method provides a robust tool for simultaneous quantitative analysis of these two chemically different groups of analytes.
Jure Zekič is a student of the doctoral program in chemical sciences at the Faculty of Chemistry and Chemical Technology, University of Ljubljana, Ljubljana, Slovenia.
Conceptualization, M.K. and J.Z.; investigation, J.Z.; supervision, M.K.; data analysis, J.Z. and M.K.; writing—original draft preparation, J.Z.; writing—review and editing, J.Z. and M.K. All authors have read and agreed to the published version of the manuscript.
Samples of the compounds are available from the authors.
This research was supported by the Slovenian Research Agency (research core funding No. P1-0005).
Conflict of interest
The authors declare no conflict of interest.
- Abuhasira, R.; Shbiro, L.; Landschaft, Y. (2018). "Medical use of cannabis and cannabinoids containing products - Regulations in Europe and North America". European Journal of Internal Medicine 49: 2–6. doi:10.1016/j.ejim.2018.01.001. PMID 29329891.
- Hutchison, K.E.; Bidwell, L.C.; Ellingson, J.M. et al. (2019). "Cannabis and Health Research: Rapid Progress Requires Innovative Research Designs". Value in Health 22 (11): 1289–94. doi:10.1016/j.jval.2019.05.005. PMID 31708066.
- Beauchet, O. (2018). "Medical cannabis use in older patients: Update on medical knowledge". Maturitas 118: 56–9. doi:10.1016/j.maturitas.2018.10.010. PMID 30415756.
- Ebbert, J.O.; Scharf, E.L.; Hurt, R.T. (2018). "Medical Cannabis". Mayo Clinic Proceedings 93: 1842–47. doi:10.1016/j.mayocp.2018.09.005. PMID 30522595.
- Blasco-Benito, S.; Seijo-Vila, M.; Caro-Villalobos, M. et al. (2018). "Appraising the "entourage effect": Antitumor action of a pure cannabinoid versus a botanical drug preparation in preclinical models of breast cancer". Biochemical Pharmacology 157: 285–93. doi:10.1016/j.bcp.2018.06.025. PMID 29940172.
- Booth, J.K.; Bohlmann, J. (2019). "Terpenes in Cannabis sativa - From plant genome to humans". Plant Science 284: 67–72. doi:10.1016/j.plantsci.2019.03.022. PMID 31084880.
- Namdar, D.; Mazuz, M.; Ion, A. et al. (2018). "Variation in the compositions of cannabinoid and terpenoids in Cannabis sativa derived from inflorescence position along the stem and extraction methods". Industrial Crops and Products 113: 376–82. doi:10.1016/j.indcrop.2018.01.060.
- Delgado-Povedano, M.M.; Callado, C.S.-C.; Priego, Capote, F. et al. (2020). "Untargeted characterization of extracts from Cannabis sativa L. cultivars by gas and liquid chromatography coupled to mass spectrometry in high resolution mode". Talanta 208: 120384. doi:10.1016/j.talanta.2019.120384.
- Nuutinen, T. (2018). "Medicinal properties of terpenes found in Cannabis sativa and Humulus lupulus". European Journal of Medicinal Chemistry 157: 198–228. doi:10.1016/j.ejmech.2018.07.076.
- Citti, C.; Linciano, P.; Russo, F. et al. (2019). "A novel phytocannabinoid isolated from Cannabis sativa L. with an in vivo cannabimimetic activity higher than Δ9-tetrahydrocannabinol: Δ9-Tetrahydrocannabiphorol". Scientific Reports 9: 20335. doi:10.1038/s41598-019-56785-1.
- Gambaro, V.E.; Froldi, R.; Saligari, E. et al. (1995). "A simple method for analyzing cannabis by gas chromatography". Acta Toxicologica Argentina 3: 11–13.
- Gibson, C.R.; Williams, R.D.; Browder, R.O. (1998). "Analysis of Hempen Ale for cannabinoids". Journal of Analytical Toxicology 22 (2): 179. doi:10.1093/jat/22.2.179. PMID 9547418.
- Ross, S.A.; Mehmedic, Z.; Murphy, T.P. et al. (2008). "GC-MS analysis of the total delta9-THC content of both drug- and fiber-type cannabis seeds". Journal of Analytical Toxicology 24 (8): 715–7. doi:10.1093/jat/24.8.715. PMID 11110027.
- Raharjo, T.J.; Verpoorte, R. (2004). "Methods for the analysis of cannabinoids in biological materials: A review". Phytochemical Analysis 15 (2): 79–94. doi:10.1002/pca.753. PMID 15116938.
- Pellegrini, M.; Marchei, E.; Pacifici, R. et al. (2005). "A rapid and simple procedure for the determination of cannabinoids in hemp food products by gas chromatography-mass spectrometry". Journal of Pharmaceutical and Biomedical Analysis 36 (5): 939–46. doi:10.1016/j.jpba.2004.07.035.
- Gambaro, V.; Dell'Acqua, L.; Farè, F. et al. (2002). "Determination of primary active constituents in Cannabis preparations by high-resolution gas chromatography/flame ionization detection and high-performance liquid chromatography/UV detection". Analtica Chimica Acta 468 (2): 245-254. doi:10.1016/S0003-2670(02)00660-8.
- Cardenia, V.; Toschi, T.G.; Scappini, S. et al. (2018). "Development and validation of a Fast gas chromatography/mass spectrometry method for the determination of cannabinoids in Cannabis sativa L". Journal of Food and Drug Analysis 26 (4): 1283–92. doi:10.1016/j.jfda.2018.06.001.
- United Nations Office on Drug and Crime (2009) (PDF). Recommended methods for the identification and analysis of cannabis and cannabis products. United Nations. ISBN 9789211482423. https://www.unodc.org/documents/scientific/ST-NAR-40-Ebook_1.pdf. Retrieved 03 April 2020.
- De Backer, B.; Debrus, B.; Lebrun, P. et al. (2009). "Innovative development and validation of an HPLC/DAD method for the qualitative and quantitative determination of major cannabinoids in cannabis plant material". Journal of Chromatography B 877 (32): 4115-4124. doi:10.1016/j.jchromb.2009.11.004.
- Aizpurua-Olaizola, O.; Omar, J.; Navarro, P. et al. (2014). "Identification and quantification of cannabinoids in Cannabis sativa L. plants by high performance liquid chromatography-mass spectrometry". Analytical and Bioanalytical Chemistry 406: 7549–60. doi:10.1007/s00216-014-8177-x.
- Zgair, A.; Wong, J.C.M.; Sabri, A. et al. (2015). "Development of a simple and sensitive HPLC–UV method for the simultaneous determination of cannabidiol and Δ9-tetrahydrocannabinol in rat plasma". Journal of Pharmaceutical and Biomedical Analysis 114: 145–51. doi:10.1016/j.jpba.2015.05.019.
- Gul, W.; Gul, S.W.; Radwan, M.M. et al. (2015). "Determination of 11 Cannabinoids in Biomass and Extracts of Different Varieties of Cannabis Using High-Performance Liquid Chromatography". Journal of AOAC International 98 (6): 1523–8. doi:10.5740/jaoacint.15-095. PMID 26651563.
- Giese, M.W.; Lewis, M.A.; Giese, L. et al. (2015). "Development and Validation of a Reliable and Robust Method for the Analysis of Cannabinoids and Terpenes in Cannabis". Journal of AOAC International 98 (6): 1503–22. doi:10.5740/jaoacint.15-116. PMID 26651562.
- Citti, C.; Ciccarella, G.; Braghiroli, D. et al. (2016). "Medicinal cannabis: Principal cannabinoids concentration and their stability evaluated by a high performance liquid chromatography coupled to diode array and quadrupole time of flight mass spectrometry method". Journal of Pharmaceutical and Biomedical Analysis 128: 201–9. doi:10.1016/j.jpba.2016.05.033. PMID 27268223.
- Patel, B.; Wene, D.; Fan, Z.T. (2017). "Qualitative and quantitative measurement of cannabinoids in cannabis using modified HPLC/DAD method". Journal of Pharmaceutical and Biomedical Analysis 146: 15–23. doi:10.1016/j.jpba.2017.07.021.
- Ciolino, L.A.; Ranieri, T.L.; Taylor, A.M. (2018). "Commercial cannabis consumer products part 2: HPLC-DAD quantitative analysis of cannabis cannabinoids". Forensic Science International 289: 438–47. doi:10.1016/j.forsciint.2018.05.033.
- Mudge, E.M.; Murch, S.J.; Brown, P.N. (2017). "Leaner and greener analysis of cannabinoids". Analytical and Bioanalytical Chemistry 409: 3153–63. doi:10.1007/s00216-017-0256-3.
- Wang, M.; Wang, Y.-H.; Avula, B. et al. (2016). "Decarboxylation Study of Acidic Cannabinoids: A Novel Approach Using Ultra-High-Performance Supercritical Fluid Chromatography/Photodiode Array-Mass Spectrometry". Cannabis and Cannabinoid Research 1 (1): 262-271. doi:10.1089/can.2016.0020.
- Dussy, F.E.; Hamberg, C.; Luginbühl, M. et al. (2005). "Isolation of Δ9-THCA-A from hemp and analytical aspects concerning the determination of Δ9-THC in cannabis products". Forensic Science International 149 (1): 3–10. doi:10.1016/j.forsciint.2004.05.015.
- Kemp, P.M.; Abukhalaf, I.K.; Manno, J.E. et al. (2005). "Cannabinoids in Humans. I. Analysis of Δ9-Tetrahydrocannabinol and Six Metabolites in Plasma and Urine Using GC-MS". Journal of Analytical Toxicology 19 (5): 285–91. doi:10.1093/jat/19.5.285.
- Steinmeyer, S.; Bregel, D.; Warth, S. et al. (2002). "Improved and validated method for the determination of Δ9-tetrahydrocannabinol (THC), 11-hydroxy-THC and 11-nor-9-carboxy-THC in serum, and in human liver microsomal preparations using gas chromatography–mass spectrometry". Journal of Chromatography B 772 (2): 239–48. doi:10.1016/S1570-0232(02)00102-2.
- Leghissa, A.; Hildenbrand, Z.L.; Foss, F.W. et al. (2018). "Determination of cannabinoids from a surrogate hops matrix using multiple reaction monitoring gas chromatography with triple quadrupole mass spectrometry". Journal of Separation Science 41 (2): 459–68. doi:10.1002/jssc.201700946.
- Križman, M. (2019). "A simplified approach for isocratic HPLC analysis of cannabinoids by fine tuning chromatographic selectivity". European Food Research and Technology 246: 315–22. doi:10.1007/s00217-019-03344-7.
- Phenomenex. "Terpenes and Cannabinoids in Marijuana by GC/FID on ZB-35HT". Application Search by Name. https://www.phenomenex.com/Application/Detail/22828. Retrieved 18 March 2020.
- Restek. "Medical Cannabis Cannabinoids and Internal Standards on Rxi-35Sil MS". Restek Searchable Chromatogram Library. https://www.restek.com/chromatogram/view/GC_FS0549/cannabinoids. Retrieved 18 March 2020.
- Nissen, L.; Zatta, A.; Stefanini, I. et al. (2010). "Characterization and antimicrobial activity of essential oils of industrial hemp varieties (Cannabis sativa L.)". Fitoterapia 81 (5): 413–9. doi:10.1016/j.fitote.2009.11.010. PMID 19969046.
- Russo, E.B. (2011). "Taming THC: potential cannabis synergy and phytocannabinoid-terpenoid entourage effects". British Journal of Pharmacology 163 (7): 1344–64. doi:10.1111/j.1476-5381.2011.01238.x. PMC PMC3165946. PMID 21749363. http://www.pubmedcentral.nih.gov/articlerender.fcgi?tool=pmcentrez&artid=PMC3165946.
- Osman, A.; Thorpe, J.W.; Caddy, B. (1985). "Comparison of Cannabis Samples from Different Origins by the Headspace Technique and an Assessment of Chromatographic Traces Using the r-Matrix". Journal of the Forensic Science Society 25 (6): 427–33. doi:10.1016/S0015-7368(85)72431-0.
- Franchina, F.A.; Dubois, L.M.; Focant, J.-F. (2020). "In-Depth Cannabis Multiclass Metabolite Profiling Using Sorptive Extraction and Multidimensional Gas Chromatography with Low- and High-Resolution Mass Spectrometry". Analytical Chemistry 92 (15): 10512-10520. doi:10.1021/acs.analchem.0c01301. PMID 32602704.
- Adams, D.J.; Dyson, P.J.; Tavener, S.J. (2004). Chemistry in Alternative Reaction Media. Wiley. p. 4. ISBN 0471498491.
- Restek. "Terpenes in Blueberry Jack Medical Cannabis on Rxi-624Sil MS". Restek Searchable Chromatogram Library. https://www.restek.com/chromatogram/view/GC_FS0515. Retrieved 07 April 2020.
- Restek. "Cannabis Terpenes Standard #1 & #2 on Rxi-624Sil MS". Restek Searchable Chromatogram Library. https://www.restek.com/chromatogram/view/GC_GN1200. Retrieved 07 April 2020.
- Fausett, A. (26 May 2020). "Analysis of Terpene and Terpenoid Content in Cannabis Sativa Using Headspace with GC/MSD" (PDF). Agilent Technologies, Inc. https://www.agilent.com/cs/library/applications/application-terpenes-8890-gcms-headspace-5994-1497en-agilent.pdf. Retrieved 07 April 2020.
This presentation is faithful to the original, with only a few minor changes to presentation. Some grammar and punctuation was cleaned up to improve readability. In some cases important information was missing from the references, and that information was added.